INSTRUCTIONS & STUDIES
This link provides "DFU" Directions For Use booklets for the current model Autoscaler® device.
This link provides Build-In Instructions for the current model Autoscaler® device.
Infection Control Instructions
Ultrasonic scaling inserts are furnished non-sterile and require sterilization before initial use and after each use thereafter. Please heed to the manufacturer’s directions for all sterilization and usage requirements.
This information supplements government agency guidelines for reducing cross contamination of infectious diseases during routine dental care. In the event any regulatory agency disagrees with this information, the agency requirements take precedence.
Before placing ultrasonic inserts into the handpiece, performing the following steps will increase the life of the o-ring on inserts:
Fill entire handpiece with water.
Lubricate o-ring (black or green rubber material) with water.
Gently tum the insert until the o-ring snaps into place.
Study 2: Dimensions of Ultrasonic Therapy Dr. Samuel B. Low DDS, MD, Med Publications, Inc. Copyright (C) 2005 Belmont Publications, Inc. All Rights Reserved.